Application of Impulse Iq Adjusting Instrument
Suitable for physical massage, used to relieve muscle pain, relax
muscles and relieve muscle fatigue.
Working principle of Impulse Iq Adjusting Instrument :
The Impulse Iq Adjusting Instrument generates a time-varying
magnetic field by pressurizing the coil with high-voltage pulses,
and uses the impact force generated by the core vibration vibration
head to treat human lesions.
The electric muscle vibrator sets the corresponding frequency,
strength, frequency limit or automatic mode through the host to
control the vibration gun, and then the vibration gun performs data
processing, and controls the corresponding unit to perform the
Main performance index of Impulse Iq Adjusting Instrument
|Impact||The impact force can be adjusted in five levels. The maximum impact
force of each level is: Level I: 100N, Level II: 200N, Level III:
300N, Level IV: 400N, Level V: 500N, tolerance ±50N|
|Output frequency||The output frequency can be adjusted in six levels, the frequency
of level I: 2Hz, the frequency of level II: 4Hz, the frequency of
level III: 6Hz, the frequency of level IV: 8Hz, the frequency of
level V: 10Hz, the frequency of level VI: 14Hz, the tolerance is
|Hits||When the automatic mode is not turned on, the number of hits is
divided into unlimited times, 5 times, 10 times, 15 times, 20
times, 25 times, 30 times, 35 times, 40 times, 45 times, 50 times,
a total of 11 levels adjustable; automatic mode Down, the number of
hits each time appears randomly in the range of 10 to 30 times|
|Noise||When the product is working normally, the noise is not more than
Contraindications of Impulse Iq Adjusting Instrument
- Unhealed part of fracture
- Bone cannot bear weight due to pathological process
- Infections or ulcers on the body surface
- Pregnant women, cancer, tuberculosis patients
- Local skin damage, skin allergies and people with bleeding tendency
Pictures of Impulse Iq Adjusting Instrument
Accessory loading and unloading
1) Accessory connection
(1) Connect the vibration gun and the host, and select the
corresponding vibration head as required;
(2) Connect the electric muscle vibrator to the AC220V, 50Hz power
socket through the power cord.
2) Disassembly of accessories
(1) Remove the power cord from the electric muscle vibrator and the
(2) Remove the vibrating gun from the host and take off the
During transportation, avoid rain and snow splash, vibration and
The packaged product should be stored in a clean room with a
relative humidity of no more than 85%, a temperature of -10°C to
40°C, no corrosive odor, and a well-ventilated, clean room.
1) Please clean the product immediately after each use, wipe it
with a slightly damp towel, and dry it naturally.
2) Put the product in a cool, dry environment, not place it nearby
or near a water source, or expose the product to direct sunlight.
3) Turn off the power switch of the product when cleaning the
4) A plug with a grounding pin must be used when using the product.
5) Ensure that the product voltage is appropriate. If the local
power supply voltage is inconsistent, it is recommended that the
user add a regulated power supply with matching power.
6) Do not bring the product close to a strong heat source, as this
may affect the life and normal use of the product.
7) Please turn off the power switch of the product when the product
is not in use, and ensure that the main power supply of the product
is turned off after use to ensure the safety of the electrical
8) Do not install or operate the product near water sources or when
your hands are wet. Be careful not to spill liquid on the product.
9) Do not insert any type of objects into the opening of the
product, otherwise there may be fire or electric shock.
10) Do not place the product on an unstable cart, shelf or cart. If
it falls over, the product may be seriously damaged.
In the short-term final processing of the product and its related
accessories, it is necessary to consider the small amount of lead
in the solder, and the environmental protection policy of the local
government should be followed when processing, and it must not be
replaced at will.